We carried out static fatigue tests on the component assembled over the implant, simulating the most critical condition, defined by the component with the highest trans-mucosal height and lower bulkiness (Arcsys 3x6x5.5mm Foldable Abutment already angled to 20°). This test was conducted according to the standard ABNT NBR ISO 14801:2012, which defines a standardized method for the verification of mechanical resistance of components under static fatigue (compression test) and dynamic fatigue. For that test, the component resisted to an average value of 1725N, or rather, a value much higher that the one from the study carried out by Ferrario et al (2004), which showed that the average maximum force applied to a single tooth (first molar) of a healty adult male is, approximately, 306N.
https://www.fgm.ind.br/wp-content/uploads/2020/10/Logo-FGM.png 0 0 comunicacao https://www.fgm.ind.br/wp-content/uploads/2020/10/Logo-FGM.png comunicacao2019-09-17 00:00:002019-09-17 00:00:00What is the maximum force (Ncm) that the component stands?
Was this answer helpful ? Yes (0) / No (0)